High-Tech Life X-ray Sensor From UK
|FOB Unit Price:||US $1,900 US $1,700|
|Purchase Qty. (Pieces)||FOB Unit Price|
|Production Capacity:||10000PCS Per Year|
|Payment Terms:||T/T, Western Union, Paypal, Money Gram|
- Model NO.: OSA-F062UK
- External Dimension (mm): 40*26
- Sensor Thickness: 6mm
- Power: 5V±0.5V
- Cable Length: >3m
- Transport Package: Box
- Origin: UK
- Type: X-Ray Equipment
- Sensoractivearea: 30*22.5
- Dynamic Range: 0~4
- Image Transfer: USB2.0
- Trademark: ATECO
- Specification: CE ISO
- HS Code: 90184990
The sensor and control box is connected already, please check the light on control box,
before taking x-ray image. The power indicator on the control box displays yellow
light and working indicator flash green light.
The sensor active surface is flat including the size 1 and size 2.
*Size1, universal sensor---Use for regular procedures, both for children and adult .
*Size2 sensor---Use for bitewings procedures.
The sensor non-reactive to X-Rays surface, contains the cable attachment.
1 Sensor non-reactive to X-Rays surface
2 Sensor active surface
1.12 Control Box
The sensor control box contains all the electronics of the sensor. The two LED light
shows whether the sensor is workable.
1.2 Technical Specifications
Sensor:APS CMOS sensor
External dimension (mm) :44×32() ; 40×26(AT-301)
SensorActiveArea : 36×27() ; 30×22.5(AT-301)
Sensor Thickness : 6mm
Dynamic Range : 0~4,096
Power : 5V±0.5V
Image Transfer :USB2.0
Cable Length :.≥ 3m
1.3 Sharing the Sensor Between Rooms
You can share the sensor between several rooms to provide access for several dentist
based on ATECO Server software. The computer must have the ATECO Dentist dental
imaging software, and ATECO Server must installed in one computer.
To share the sensor between several computers, move it from room to room. When
you connect the sensor to a USB 2.0 port on the computer, the sensor is recognized
automatically and is operational.
To share data between rooms, you can connect them to remote data---ATECO Server
database. The ATECO Dentist dental imaging software needs only to access a shared
database on the same computer or on a remote computer, which means ATECO Server
1.4 Using the different Positioning Systems
There are two ways to position the sensor in the patient mouth to get an classic
radiology. You may spend some time to adapt due to the rigidity of the sensor.
One method is angular bisector technique, the other method is paralleling technique. It
is just the way to position sensor, which can be chosen by practitioner's experience.
X-Ray Generator Compatibility
Normally, the sensor is compatible with all generators which meets the present
standard of intraoral radiology. You can use a high frequency or conventional
generator. To achieve gaining better images, the generator must operate with a voltage
of 65 to 70kV.
1As a precision devices, avoid flop,pulling and long-time disinfectant soak
2.Put disposable plastic jackets before used, avoid cross infection and allergy
3.No bite, for fear that the sensor or the jacket been broken
4.Caution for epileptic or psychopath
5.Learn the user manual carefully before use
6.User should be the professional dental or technician
2. Imaging Software General Introduction
2.1 Computer System Requirements
Processor: Intel 1.2GHz chip or above;
Memory: Above 1G ;
Hard disk:Above 40G;
Interface: USB 2.0;
Display: Resolution 1024 × 758 (15") or above
Operating System: Windows XP
The computer connected to system shall be in accordance with IEC 60950-1:6005.
2.2 Imaging Software
TheAteco AT-301 dental imaging system operates with the following software:
*Atecotech Server for sharing information between workstations.
Atecotech Software is a user-friendly working interface that was designed and
developed specifically for radiological diagnosis. It is the common imaging platform
for all our digital systems for dentistry.